The stressors will allow specific stimulations of the cardio vascular system.
|CARDIOLAB Elements||Operational mode||Acronym||Comments|
|Cold/Warm pressure glove||Stationary||CWPG||It shall allow to apply thermal stress on the fore arm in a range from -0°C to 45°C with a regulation in the case of positive temperature of 0.5°C.|
|Leg/Arm cuff system||Stationary||LACS||It allows to apply at the level of the limbs occlusive stress in a range from 0mm of Hg to 300mm of Hg. The accuracy is about 1mm of Hg and the deflation speed has to be improved to TTBBDD mm of Hg per second. It allows to apply two different level/profile of pressure on the arms and on the legs.|